iLIGHT® Essential Hair Removal System
The Remington iLIGHT Essential is a light-based device for permanent hair reduction designed for home use. It is a powerful electrical device and should be used with special attention to safety. Be sure to read all warnings, contraindications and safety precautions before use, and follow them strictly. We recommend you re-familiarize yourself with the instructions before each treatment.
Indications for use
This device is an over-the-counter device intended to remove unwanted hair and effect long-term, or permanent hair reduction. Permanent hair reduction is defined as a long-term, stable reduction in the number of hairs regrowing when measured at 6, 9 and 12 months after the completion of a treatment regimen.
You may use the device to remove unwanted body hair. Ideal body areas include the underarms, bikini line, arms, legs, face, back, shoulders and chest. Do not use this device for any purpose other than as described.
With your device, safety comes first. Home Pulsed Light (HPL) technology in your device provides extra safety with lower energy levels. HPL achieves long-term hair removal results at a fraction of the energy level used in other light-based hair removal equipment. The HPL technology reduces the risk of harm or complications, and contributes to your overall safety. Your device has additional features to protect your eyes and your skin.
Eyes: Light pulses are only active when the treatment window is in full contact with the skin. Nevertheless, do not look at the treatment window when you are using the device. This may cause damage to the eyes.
Skin: Your skin is never in direct contact with the optical lens through which light pulses are emitted.
LONG-TERM HAIR REMOVAL
The efficiency of hair removal varies from person to person according to body area, hair color, and how the device is used. A typical full hair growth cycle may take 18-24 months. During this time, multiple treatment sessions are required to achieve permanent hair removal.
A typical device hair removal plan during a full hair growth cycle includes:
Treatments 1-4: Plan two weeks apart.
Treatments 5-7: Plan four weeks apart.
Treatments 8+: Treat as needed, until desired results are achieved
Model no. IPL3500
Spot size 0.9cm x 3cm [2.7cm²]
Technology Home Pulsed Light
Maximum energy level Max 4J/cm²
Operation & safety Skin tone tester ensures use on appropriate skin types.
Package size 180 x 180 x 100 mm
System weight 200 gm
Transport & storage between uses and storage condition
Temperature: -40°C to +70°C
Relative humidity : 10% to 90%
Atmospheric pressure: 500 to 1060 hPa
Temperature: 10°C to 35°C
Relative humidity: 30% to 75%
Atmospheric pressure: 700 to 1060 hPa
Time from minimum storage temperature between uses to minimum operating temperature with ambient temperature of 20°C degrees
30 minutes Time from maximum storage temperature between uses to maximum operating temperature with ambient temperature of 20°C degrees
30 minutes Mode of operation Continuous
Adapter models and rating
KSA24A120015HE Input: 100-240VAC,
50-60 Hz, 0.5A
PMA18E-120 Input: 100-240VAC,
50-60 Hz, 0.6A
Output: 12V, 1.5A
Protection against electrical shock Class II Medical equipment type BF
This device was tested and complies with:
EC 60601-1 (Safety)
IEC 60601-1-2 (EMC)
IEC 60601-1-11 (Environment)
IEC60601-2-57 (Safety of non-laser source)
Expected service life of the equipment 5 years